Model No.: DSW 905B905B, 906B
|Type||Brand Name||Model Number||Place of Origin||Applicable Gas||Input Pressure||Output Pressure||Safety valve||The Range of Flow|
|Breathing Apparatus||DSW||DSW 905B905B,906B||DSW China||Oxygen||15MPa||0.2-0.3MPa||0.35±0.05MPa||1-15L/min|
|Connection Thread||Structure||Wet the cup||Humidity Conditioning||Packaging Details||Dimension||G.W||N.W||Material|
hot sale female medical oxygen regulator
Outside thread G5/8,
Flow 0-10L/min or 0-15L/min
Output pressure 0.2-0.3MPa,
Input pressure 150bar
gas inlet types:
1.gas -specific inlet probe conforming to BS ,AMERICAN, GERMAN, FRENCH
2.1/8'' NPT female-threadde inlet
3.Bullnose nipple with 3/8'' BSPNUT. Other country-specific direct and remote-fitting configurations are
available- contact the office.
gas outlet types:
1.body has female thread -use BMI 'universal'tubing nipple
2.body has a 9/16'' UNFmale thread -use tubing nipple(oxygen)
1.direct-probe fits into wall-mounted gas -specific socket(antirotation device maintains flowmeter upright)
2.remote-kit available with 2m colour-coded flexible hose,and a gas-specific connector the flowmeter is mounted
remotely from the gas source on:
NOTE:the rail-mounting kit utilises a "universal' rail clamp,allowing attachment to BMI,ohmeda,vickers"modura",
din rail section.
operation and use
2.service & maintence
do dot disassemble when connected to a source of pressure
repairs must only be undertaken by suitably trained personnel.
the pre-use checks in section 2, and the leakage check in section 3.2 must be carried out after any disassembly
if the cover tube is cracked, withdraw the unit from service and renew:
1.disconnect the flowmeter from the pressure source
2.open the flow control valve to relieve any contained pressure.
3.hold the unit vertically (to retain the flow tube and ball )unscrew the cover tube and lift off.
calibrated tube and indicator ball
4.lift off the flow tube , taking care not to let the ball roll out
5.remove the tube connector adaptor and the cover tube o-ring
6.fit a new cover tube o-ring at the base of the body.
7.fit a new tube adaptor. Check orientation,as illustrated.
8.fit the flow tube , and ball over the adaptor.
9.position the tube with the scale towards the front of the unit.
10.fit the outer tube,and screw clockwise into the flowmeter body.
4. environmental conditions
store/operate between 5 and 25 degree
protect exposed o-rings from light
Contact us if you need more details on Oxygen Regulator. We are ready to answer your questions on packaging, logistics, certification or any other aspects about Oxygen Inhaler、Inhaler. If these products fail to match your need, please contact us and we would like to provide relevant information.
Product Categories : Medical Oxygen Gas Cylinder
Premium Related Products